Enter search content
Ordinary Gloss Plates
★ For the verification of gloss work machine.
★ GS-G type is glass material board; GS-C type is ceramic material board;
★ Specifications: 45mm & times; 85mm
★ annual variation &&;;plusmn;0.5 gloss unit
Note: The working surface of theGS_G type has an interference aperture of no more than 15 apertures and a partial aperture of no more than 3.
★ The entire set of instruments should be kept clean during operation. Failure to do so may result in measurement errors and inaccurate verification.
★ If there is fog, dust or smudge on the surface of the work board, please wipe it off with lens paper or soft cloth and anhydrous ethanol.
★The instrument measuring port should be attached to the measured surface to avoid external light leakage. If necessary, black light cloth and other objects can be used to block the light around the measuring mouth. It is recommended to avoid glare environmental measurements.
★ Calibration and measurement or calibration operations should be performed at the same ambient temperature, otherwise too large ambient temperature differences will result in digital offsets.
★ If the MG6 series gloss meter is not used for a long time, it should be withdrawn from the battery to avoid corrosion and damage to the instrument.
★When the WGG60 series gloss meter is not used for a long time, it should replenish the battery pack in the machine every month.
After eighteen-years’development, KSJ Glossmeters have been widely used all over the world in gloss measuring of paintwork, decorative materials, woodenwares, ceramic, printing ink, paper, as well as metal polishing and depositing. And becoming the first choice for experts in the above fields.